[Hallb-engineering] DC Mix 2 CO2 Flow
beng at jlab.org
Sat Jul 28 14:27:15 EDT 2018
The regulator for the CO2 used for Mix 2 needs to be adjusted.
The MFC valve was at 100% but not flowing anything, which is almost always due to the upstream pressure not being high enough. In the attached plot the scale for the MFC is -2 to 2 SLM.
I put the mixing system in manual mode and set the flows for mix 2 to 0 so the tank won't be filled with just Argon.
Once the regulator is adjusted or the pressure in the tank gets below ~95 it can be put back into automatic mode.
-------------- next part --------------
A non-text attachment was scrubbed...
Name: Screen Shot 2018-07-28 at 2.24.17 PM.png
Size: 12842 bytes
Desc: not available
More information about the Hallb-engineering